#		<< IRIS SEED Reader, Release 5.4beta-USGS-20190905 >>
#		
#		======== CHANNEL RESPONSE DATA ========
B050F03     Station:     CSH
B050F16     Network:     NC
B052F03     Location:    ??
B052F04     Channel:     EHZ
B052F22     Start date:  1973,015,00:00:00
B052F23     End date:    1977,054,23:12:00
#		=======================================
#		+               +--------------------------------------------+                +
#		+               |   Response (Poles & Zeros),   CSH ch EHZ   |                +
#		+               +--------------------------------------------+                +
#		
B053F03     Transfer function type:                A [Laplace Transform (Rad/sec)]
B053F04     Stage sequence number:                 1
B053F05     Response in units lookup:              M/S - Velocity in Meters Per Second
B053F06     Response out units lookup:             V - Volts
B053F07     A0 normalization factor:               1.00285
B053F08     Normalization frequency:               10
B053F09     Number of zeroes:                      2
B053F14     Number of poles:                       2
#		Complex zeroes:
#		  i  real          imag          real_error    imag_error
B053F10-13    0  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
B053F10-13    1  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
#		Complex poles:
#		  i  real          imag          real_error    imag_error
B053F15-18    0 -5.026500E+00  3.769870E+00  0.000000E+00  0.000000E+00
B053F15-18    1 -5.026500E+00 -3.769870E+00  0.000000E+00  0.000000E+00
#		
#		+                  +---------------------------------------+                  +
#		+                  |       Channel Gain,   CSH ch EHZ      |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 1
B058F04     Gain:                                  1.000000E+02
B058F05     Frequency of gain:                     1.000000E+01 HZ
B058F06     Number of calibrations:                0
#		
#		+               +--------------------------------------------+                +
#		+               |   Response (Poles & Zeros),   CSH ch EHZ   |                +
#		+               +--------------------------------------------+                +
#		
B053F03     Transfer function type:                A [Laplace Transform (Rad/sec)]
B053F04     Stage sequence number:                 2
B053F05     Response in units lookup:              V - Volts
B053F06     Response out units lookup:             V - Volts
B053F07     A0 normalization factor:               1.01498E+14
B053F08     Normalization frequency:               10
B053F09     Number of zeroes:                      2
B053F14     Number of poles:                       8
#		Complex zeroes:
#		  i  real          imag          real_error    imag_error
B053F10-13    0  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
B053F10-13    1  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
#		Complex poles:
#		  i  real          imag          real_error    imag_error
B053F15-18    0 -5.969000E-01  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    1 -5.969000E-01  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    2 -2.764600E+02  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    3 -2.764600E+02  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    4 -7.213730E+01  1.741510E+02  0.000000E+00  0.000000E+00
B053F15-18    5 -7.213730E+01 -1.741510E+02  0.000000E+00  0.000000E+00
B053F15-18    6 -1.741510E+02  7.213730E+01  0.000000E+00  0.000000E+00
B053F15-18    7 -1.741510E+02 -7.213730E+01  0.000000E+00  0.000000E+00
#		
#		+                  +---------------------------------------+                  +
#		+                  |       Channel Gain,   CSH ch EHZ      |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 2
B058F04     Gain:                                  2.433090E+03
B058F05     Frequency of gain:                     5.000000E+00 HZ
B058F06     Number of calibrations:                0
#		
#		+                  +---------------------------------------+                  +
#		+                  |   Channel Sensitivity,   CSH ch EHZ   |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 0
B058F04     Sensitivity:                           2.433090E+05
B058F05     Frequency of sensitivity:              1.000000E+01 HZ
B058F06     Number of calibrations:                0
#		
#		<< IRIS SEED Reader, Release 5.4beta-USGS-20190905 >>
#		
#		======== CHANNEL RESPONSE DATA ========
B050F03     Station:     CSH
B050F16     Network:     NC
B052F03     Location:    ??
B052F04     Channel:     EHZ
B052F22     Start date:  1977,054,23:12:00
B052F23     End date:    1977,234,16:30:00
#		=======================================
#		+               +--------------------------------------------+                +
#		+               |   Response (Poles & Zeros),   CSH ch EHZ   |                +
#		+               +--------------------------------------------+                +
#		
B053F03     Transfer function type:                A [Laplace Transform (Rad/sec)]
B053F04     Stage sequence number:                 1
B053F05     Response in units lookup:              M/S - Velocity in Meters Per Second
B053F06     Response out units lookup:             V - Volts
B053F07     A0 normalization factor:               1.00285
B053F08     Normalization frequency:               10
B053F09     Number of zeroes:                      2
B053F14     Number of poles:                       2
#		Complex zeroes:
#		  i  real          imag          real_error    imag_error
B053F10-13    0  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
B053F10-13    1  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
#		Complex poles:
#		  i  real          imag          real_error    imag_error
B053F15-18    0 -5.026500E+00  3.769870E+00  0.000000E+00  0.000000E+00
B053F15-18    1 -5.026500E+00 -3.769870E+00  0.000000E+00  0.000000E+00
#		
#		+                  +---------------------------------------+                  +
#		+                  |       Channel Gain,   CSH ch EHZ      |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 1
B058F04     Gain:                                  1.000000E+02
B058F05     Frequency of gain:                     1.000000E+01 HZ
B058F06     Number of calibrations:                0
#		
#		+               +--------------------------------------------+                +
#		+               |   Response (Poles & Zeros),   CSH ch EHZ   |                +
#		+               +--------------------------------------------+                +
#		
B053F03     Transfer function type:                A [Laplace Transform (Rad/sec)]
B053F04     Stage sequence number:                 2
B053F05     Response in units lookup:              V - Volts
B053F06     Response out units lookup:             V - Volts
B053F07     A0 normalization factor:               1.01498E+14
B053F08     Normalization frequency:               10
B053F09     Number of zeroes:                      2
B053F14     Number of poles:                       8
#		Complex zeroes:
#		  i  real          imag          real_error    imag_error
B053F10-13    0  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
B053F10-13    1  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
#		Complex poles:
#		  i  real          imag          real_error    imag_error
B053F15-18    0 -5.969000E-01  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    1 -5.969000E-01  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    2 -2.764600E+02  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    3 -2.764600E+02  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    4 -7.213730E+01  1.741510E+02  0.000000E+00  0.000000E+00
B053F15-18    5 -7.213730E+01 -1.741510E+02  0.000000E+00  0.000000E+00
B053F15-18    6 -1.741510E+02  7.213730E+01  0.000000E+00  0.000000E+00
B053F15-18    7 -1.741510E+02 -7.213730E+01  0.000000E+00  0.000000E+00
#		
#		+                  +---------------------------------------+                  +
#		+                  |       Channel Gain,   CSH ch EHZ      |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 2
B058F04     Gain:                                  2.433090E+03
B058F05     Frequency of gain:                     5.000000E+00 HZ
B058F06     Number of calibrations:                0
#		
#		+                  +---------------------------------------+                  +
#		+                  |   Channel Sensitivity,   CSH ch EHZ   |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 0
B058F04     Sensitivity:                           2.433090E+05
B058F05     Frequency of sensitivity:              1.000000E+01 HZ
B058F06     Number of calibrations:                0
#		
#		<< IRIS SEED Reader, Release 5.4beta-USGS-20190905 >>
#		
#		======== CHANNEL RESPONSE DATA ========
B050F03     Station:     CSH
B050F16     Network:     NC
B052F03     Location:    ??
B052F04     Channel:     EHZ
B052F22     Start date:  1977,234,16:30:00
B052F23     End date:    1977,263,18:53:00
#		=======================================
#		+               +--------------------------------------------+                +
#		+               |   Response (Poles & Zeros),   CSH ch EHZ   |                +
#		+               +--------------------------------------------+                +
#		
B053F03     Transfer function type:                A [Laplace Transform (Rad/sec)]
B053F04     Stage sequence number:                 1
B053F05     Response in units lookup:              M/S - Velocity in Meters Per Second
B053F06     Response out units lookup:             V - Volts
B053F07     A0 normalization factor:               1.00285
B053F08     Normalization frequency:               10
B053F09     Number of zeroes:                      2
B053F14     Number of poles:                       2
#		Complex zeroes:
#		  i  real          imag          real_error    imag_error
B053F10-13    0  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
B053F10-13    1  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
#		Complex poles:
#		  i  real          imag          real_error    imag_error
B053F15-18    0 -5.026500E+00  3.769870E+00  0.000000E+00  0.000000E+00
B053F15-18    1 -5.026500E+00 -3.769870E+00  0.000000E+00  0.000000E+00
#		
#		+                  +---------------------------------------+                  +
#		+                  |       Channel Gain,   CSH ch EHZ      |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 1
B058F04     Gain:                                  1.000000E+02
B058F05     Frequency of gain:                     1.000000E+01 HZ
B058F06     Number of calibrations:                0
#		
#		+               +--------------------------------------------+                +
#		+               |   Response (Poles & Zeros),   CSH ch EHZ   |                +
#		+               +--------------------------------------------+                +
#		
B053F03     Transfer function type:                A [Laplace Transform (Rad/sec)]
B053F04     Stage sequence number:                 2
B053F05     Response in units lookup:              V - Volts
B053F06     Response out units lookup:             V - Volts
B053F07     A0 normalization factor:               1.01498E+14
B053F08     Normalization frequency:               10
B053F09     Number of zeroes:                      2
B053F14     Number of poles:                       8
#		Complex zeroes:
#		  i  real          imag          real_error    imag_error
B053F10-13    0  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
B053F10-13    1  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
#		Complex poles:
#		  i  real          imag          real_error    imag_error
B053F15-18    0 -5.969000E-01  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    1 -5.969000E-01  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    2 -2.764600E+02  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    3 -2.764600E+02  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    4 -7.213730E+01  1.741510E+02  0.000000E+00  0.000000E+00
B053F15-18    5 -7.213730E+01 -1.741510E+02  0.000000E+00  0.000000E+00
B053F15-18    6 -1.741510E+02  7.213730E+01  0.000000E+00  0.000000E+00
B053F15-18    7 -1.741510E+02 -7.213730E+01  0.000000E+00  0.000000E+00
#		
#		+                  +---------------------------------------+                  +
#		+                  |       Channel Gain,   CSH ch EHZ      |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 2
B058F04     Gain:                                  2.433090E+03
B058F05     Frequency of gain:                     5.000000E+00 HZ
B058F06     Number of calibrations:                0
#		
#		+                  +---------------------------------------+                  +
#		+                  |   Channel Sensitivity,   CSH ch EHZ   |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 0
B058F04     Sensitivity:                           2.433090E+05
B058F05     Frequency of sensitivity:              1.000000E+01 HZ
B058F06     Number of calibrations:                0
#		
#		<< IRIS SEED Reader, Release 5.4beta-USGS-20190905 >>
#		
#		======== CHANNEL RESPONSE DATA ========
B050F03     Station:     CSH
B050F16     Network:     NC
B052F03     Location:    ??
B052F04     Channel:     EHZ
B052F22     Start date:  1977,263,18:53:00
B052F23     End date:    1980,254,17:30:00
#		=======================================
#		+               +--------------------------------------------+                +
#		+               |   Response (Poles & Zeros),   CSH ch EHZ   |                +
#		+               +--------------------------------------------+                +
#		
B053F03     Transfer function type:                A [Laplace Transform (Rad/sec)]
B053F04     Stage sequence number:                 1
B053F05     Response in units lookup:              M/S - Velocity in Meters Per Second
B053F06     Response out units lookup:             V - Volts
B053F07     A0 normalization factor:               1.00285
B053F08     Normalization frequency:               10
B053F09     Number of zeroes:                      2
B053F14     Number of poles:                       2
#		Complex zeroes:
#		  i  real          imag          real_error    imag_error
B053F10-13    0  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
B053F10-13    1  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
#		Complex poles:
#		  i  real          imag          real_error    imag_error
B053F15-18    0 -5.026500E+00  3.769870E+00  0.000000E+00  0.000000E+00
B053F15-18    1 -5.026500E+00 -3.769870E+00  0.000000E+00  0.000000E+00
#		
#		+                  +---------------------------------------+                  +
#		+                  |       Channel Gain,   CSH ch EHZ      |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 1
B058F04     Gain:                                  1.000000E+02
B058F05     Frequency of gain:                     1.000000E+01 HZ
B058F06     Number of calibrations:                0
#		
#		+               +--------------------------------------------+                +
#		+               |   Response (Poles & Zeros),   CSH ch EHZ   |                +
#		+               +--------------------------------------------+                +
#		
B053F03     Transfer function type:                A [Laplace Transform (Rad/sec)]
B053F04     Stage sequence number:                 2
B053F05     Response in units lookup:              V - Volts
B053F06     Response out units lookup:             V - Volts
B053F07     A0 normalization factor:               1.01498E+14
B053F08     Normalization frequency:               10
B053F09     Number of zeroes:                      2
B053F14     Number of poles:                       8
#		Complex zeroes:
#		  i  real          imag          real_error    imag_error
B053F10-13    0  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
B053F10-13    1  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
#		Complex poles:
#		  i  real          imag          real_error    imag_error
B053F15-18    0 -5.969000E-01  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    1 -5.969000E-01  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    2 -2.764600E+02  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    3 -2.764600E+02  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    4 -7.213730E+01  1.741510E+02  0.000000E+00  0.000000E+00
B053F15-18    5 -7.213730E+01 -1.741510E+02  0.000000E+00  0.000000E+00
B053F15-18    6 -1.741510E+02  7.213730E+01  0.000000E+00  0.000000E+00
B053F15-18    7 -1.741510E+02 -7.213730E+01  0.000000E+00  0.000000E+00
#		
#		+                  +---------------------------------------+                  +
#		+                  |       Channel Gain,   CSH ch EHZ      |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 2
B058F04     Gain:                                  2.433090E+03
B058F05     Frequency of gain:                     5.000000E+00 HZ
B058F06     Number of calibrations:                0
#		
#		+                  +---------------------------------------+                  +
#		+                  |   Channel Sensitivity,   CSH ch EHZ   |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 0
B058F04     Sensitivity:                           2.433090E+05
B058F05     Frequency of sensitivity:              1.000000E+01 HZ
B058F06     Number of calibrations:                0
#		
#		<< IRIS SEED Reader, Release 5.4beta-USGS-20190905 >>
#		
#		======== CHANNEL RESPONSE DATA ========
B050F03     Station:     CSH
B050F16     Network:     NC
B052F03     Location:    ??
B052F04     Channel:     EHZ
B052F22     Start date:  1980,254,17:30:00
B052F23     End date:    1981,090,20:10:00
#		=======================================
#		+               +--------------------------------------------+                +
#		+               |   Response (Poles & Zeros),   CSH ch EHZ   |                +
#		+               +--------------------------------------------+                +
#		
B053F03     Transfer function type:                A [Laplace Transform (Rad/sec)]
B053F04     Stage sequence number:                 1
B053F05     Response in units lookup:              M/S - Velocity in Meters Per Second
B053F06     Response out units lookup:             V - Volts
B053F07     A0 normalization factor:               1.00285
B053F08     Normalization frequency:               10
B053F09     Number of zeroes:                      2
B053F14     Number of poles:                       2
#		Complex zeroes:
#		  i  real          imag          real_error    imag_error
B053F10-13    0  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
B053F10-13    1  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
#		Complex poles:
#		  i  real          imag          real_error    imag_error
B053F15-18    0 -5.026500E+00  3.769870E+00  0.000000E+00  0.000000E+00
B053F15-18    1 -5.026500E+00 -3.769870E+00  0.000000E+00  0.000000E+00
#		
#		+                  +---------------------------------------+                  +
#		+                  |       Channel Gain,   CSH ch EHZ      |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 1
B058F04     Gain:                                  1.000000E+02
B058F05     Frequency of gain:                     1.000000E+01 HZ
B058F06     Number of calibrations:                0
#		
#		+               +--------------------------------------------+                +
#		+               |   Response (Poles & Zeros),   CSH ch EHZ   |                +
#		+               +--------------------------------------------+                +
#		
B053F03     Transfer function type:                A [Laplace Transform (Rad/sec)]
B053F04     Stage sequence number:                 2
B053F05     Response in units lookup:              V - Volts
B053F06     Response out units lookup:             V - Volts
B053F07     A0 normalization factor:               1.01498E+14
B053F08     Normalization frequency:               10
B053F09     Number of zeroes:                      2
B053F14     Number of poles:                       8
#		Complex zeroes:
#		  i  real          imag          real_error    imag_error
B053F10-13    0  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
B053F10-13    1  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
#		Complex poles:
#		  i  real          imag          real_error    imag_error
B053F15-18    0 -5.969000E-01  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    1 -5.969000E-01  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    2 -2.764600E+02  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    3 -2.764600E+02  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    4 -7.213730E+01  1.741510E+02  0.000000E+00  0.000000E+00
B053F15-18    5 -7.213730E+01 -1.741510E+02  0.000000E+00  0.000000E+00
B053F15-18    6 -1.741510E+02  7.213730E+01  0.000000E+00  0.000000E+00
B053F15-18    7 -1.741510E+02 -7.213730E+01  0.000000E+00  0.000000E+00
#		
#		+                  +---------------------------------------+                  +
#		+                  |       Channel Gain,   CSH ch EHZ      |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 2
B058F04     Gain:                                  2.433090E+03
B058F05     Frequency of gain:                     5.000000E+00 HZ
B058F06     Number of calibrations:                0
#		
#		+                  +---------------------------------------+                  +
#		+                  |   Channel Sensitivity,   CSH ch EHZ   |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 0
B058F04     Sensitivity:                           2.433090E+05
B058F05     Frequency of sensitivity:              1.000000E+01 HZ
B058F06     Number of calibrations:                0
#		
#		<< IRIS SEED Reader, Release 5.4beta-USGS-20190905 >>
#		
#		======== CHANNEL RESPONSE DATA ========
B050F03     Station:     CSH
B050F16     Network:     NC
B052F03     Location:    ??
B052F04     Channel:     EHZ
B052F22     Start date:  1981,090,20:10:00
B052F23     End date:    1981,343,19:30:00
#		=======================================
#		+               +--------------------------------------------+                +
#		+               |   Response (Poles & Zeros),   CSH ch EHZ   |                +
#		+               +--------------------------------------------+                +
#		
B053F03     Transfer function type:                A [Laplace Transform (Rad/sec)]
B053F04     Stage sequence number:                 1
B053F05     Response in units lookup:              M/S - Velocity in Meters Per Second
B053F06     Response out units lookup:             V - Volts
B053F07     A0 normalization factor:               1.00285
B053F08     Normalization frequency:               10
B053F09     Number of zeroes:                      2
B053F14     Number of poles:                       2
#		Complex zeroes:
#		  i  real          imag          real_error    imag_error
B053F10-13    0  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
B053F10-13    1  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
#		Complex poles:
#		  i  real          imag          real_error    imag_error
B053F15-18    0 -5.026500E+00  3.769870E+00  0.000000E+00  0.000000E+00
B053F15-18    1 -5.026500E+00 -3.769870E+00  0.000000E+00  0.000000E+00
#		
#		+                  +---------------------------------------+                  +
#		+                  |       Channel Gain,   CSH ch EHZ      |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 1
B058F04     Gain:                                  1.000000E+02
B058F05     Frequency of gain:                     1.000000E+01 HZ
B058F06     Number of calibrations:                0
#		
#		+               +--------------------------------------------+                +
#		+               |   Response (Poles & Zeros),   CSH ch EHZ   |                +
#		+               +--------------------------------------------+                +
#		
B053F03     Transfer function type:                A [Laplace Transform (Rad/sec)]
B053F04     Stage sequence number:                 2
B053F05     Response in units lookup:              V - Volts
B053F06     Response out units lookup:             V - Volts
B053F07     A0 normalization factor:               1.01498E+14
B053F08     Normalization frequency:               10
B053F09     Number of zeroes:                      2
B053F14     Number of poles:                       8
#		Complex zeroes:
#		  i  real          imag          real_error    imag_error
B053F10-13    0  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
B053F10-13    1  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
#		Complex poles:
#		  i  real          imag          real_error    imag_error
B053F15-18    0 -5.969000E-01  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    1 -5.969000E-01  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    2 -2.764600E+02  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    3 -2.764600E+02  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    4 -7.213730E+01  1.741510E+02  0.000000E+00  0.000000E+00
B053F15-18    5 -7.213730E+01 -1.741510E+02  0.000000E+00  0.000000E+00
B053F15-18    6 -1.741510E+02  7.213730E+01  0.000000E+00  0.000000E+00
B053F15-18    7 -1.741510E+02 -7.213730E+01  0.000000E+00  0.000000E+00
#		
#		+                  +---------------------------------------+                  +
#		+                  |       Channel Gain,   CSH ch EHZ      |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 2
B058F04     Gain:                                  2.433090E+03
B058F05     Frequency of gain:                     5.000000E+00 HZ
B058F06     Number of calibrations:                0
#		
#		+                  +---------------------------------------+                  +
#		+                  |   Channel Sensitivity,   CSH ch EHZ   |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 0
B058F04     Sensitivity:                           2.433090E+05
B058F05     Frequency of sensitivity:              1.000000E+01 HZ
B058F06     Number of calibrations:                0
#		
#		<< IRIS SEED Reader, Release 5.4beta-USGS-20190905 >>
#		
#		======== CHANNEL RESPONSE DATA ========
B050F03     Station:     CSH
B050F16     Network:     NC
B052F03     Location:    ??
B052F04     Channel:     EHZ
B052F22     Start date:  1981,343,19:30:00
B052F23     End date:    1984,001,00:00:00
#		=======================================
#		+               +--------------------------------------------+                +
#		+               |   Response (Poles & Zeros),   CSH ch EHZ   |                +
#		+               +--------------------------------------------+                +
#		
B053F03     Transfer function type:                A [Laplace Transform (Rad/sec)]
B053F04     Stage sequence number:                 1
B053F05     Response in units lookup:              M/S - Velocity in Meters Per Second
B053F06     Response out units lookup:             V - Volts
B053F07     A0 normalization factor:               1.00285
B053F08     Normalization frequency:               10
B053F09     Number of zeroes:                      2
B053F14     Number of poles:                       2
#		Complex zeroes:
#		  i  real          imag          real_error    imag_error
B053F10-13    0  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
B053F10-13    1  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
#		Complex poles:
#		  i  real          imag          real_error    imag_error
B053F15-18    0 -5.026500E+00  3.769870E+00  0.000000E+00  0.000000E+00
B053F15-18    1 -5.026500E+00 -3.769870E+00  0.000000E+00  0.000000E+00
#		
#		+                  +---------------------------------------+                  +
#		+                  |       Channel Gain,   CSH ch EHZ      |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 1
B058F04     Gain:                                  1.000000E+02
B058F05     Frequency of gain:                     1.000000E+01 HZ
B058F06     Number of calibrations:                0
#		
#		+               +--------------------------------------------+                +
#		+               |   Response (Poles & Zeros),   CSH ch EHZ   |                +
#		+               +--------------------------------------------+                +
#		
B053F03     Transfer function type:                A [Laplace Transform (Rad/sec)]
B053F04     Stage sequence number:                 2
B053F05     Response in units lookup:              V - Volts
B053F06     Response out units lookup:             V - Volts
B053F07     A0 normalization factor:               1.01498E+14
B053F08     Normalization frequency:               10
B053F09     Number of zeroes:                      2
B053F14     Number of poles:                       8
#		Complex zeroes:
#		  i  real          imag          real_error    imag_error
B053F10-13    0  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
B053F10-13    1  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
#		Complex poles:
#		  i  real          imag          real_error    imag_error
B053F15-18    0 -5.969000E-01  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    1 -5.969000E-01  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    2 -2.764600E+02  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    3 -2.764600E+02  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    4 -7.213730E+01  1.741510E+02  0.000000E+00  0.000000E+00
B053F15-18    5 -7.213730E+01 -1.741510E+02  0.000000E+00  0.000000E+00
B053F15-18    6 -1.741510E+02  7.213730E+01  0.000000E+00  0.000000E+00
B053F15-18    7 -1.741510E+02 -7.213730E+01  0.000000E+00  0.000000E+00
#		
#		+                  +---------------------------------------+                  +
#		+                  |       Channel Gain,   CSH ch EHZ      |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 2
B058F04     Gain:                                  2.433090E+03
B058F05     Frequency of gain:                     5.000000E+00 HZ
B058F06     Number of calibrations:                0
#		
#		+                  +---------------------------------------+                  +
#		+                  |   Channel Sensitivity,   CSH ch EHZ   |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 0
B058F04     Sensitivity:                           2.433090E+05
B058F05     Frequency of sensitivity:              1.000000E+01 HZ
B058F06     Number of calibrations:                0
#		
#		<< IRIS SEED Reader, Release 5.4beta-USGS-20190905 >>
#		
#		======== CHANNEL RESPONSE DATA ========
B050F03     Station:     CSH
B050F16     Network:     NC
B052F03     Location:    ??
B052F04     Channel:     EHZ
B052F22     Start date:  1984,001,00:00:00
B052F23     End date:    1985,295,20:15:00
#		=======================================
#		+               +--------------------------------------------+                +
#		+               |   Response (Poles & Zeros),   CSH ch EHZ   |                +
#		+               +--------------------------------------------+                +
#		
B053F03     Transfer function type:                A [Laplace Transform (Rad/sec)]
B053F04     Stage sequence number:                 1
B053F05     Response in units lookup:              M/S - Velocity in Meters Per Second
B053F06     Response out units lookup:             V - Volts
B053F07     A0 normalization factor:               1.00285
B053F08     Normalization frequency:               10
B053F09     Number of zeroes:                      2
B053F14     Number of poles:                       2
#		Complex zeroes:
#		  i  real          imag          real_error    imag_error
B053F10-13    0  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
B053F10-13    1  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
#		Complex poles:
#		  i  real          imag          real_error    imag_error
B053F15-18    0 -5.026500E+00  3.769870E+00  0.000000E+00  0.000000E+00
B053F15-18    1 -5.026500E+00 -3.769870E+00  0.000000E+00  0.000000E+00
#		
#		+                  +---------------------------------------+                  +
#		+                  |       Channel Gain,   CSH ch EHZ      |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 1
B058F04     Gain:                                  1.000000E+02
B058F05     Frequency of gain:                     1.000000E+01 HZ
B058F06     Number of calibrations:                0
#		
#		+               +--------------------------------------------+                +
#		+               |   Response (Poles & Zeros),   CSH ch EHZ   |                +
#		+               +--------------------------------------------+                +
#		
B053F03     Transfer function type:                A [Laplace Transform (Rad/sec)]
B053F04     Stage sequence number:                 2
B053F05     Response in units lookup:              V - Volts
B053F06     Response out units lookup:             V - Volts
B053F07     A0 normalization factor:               1.01498E+14
B053F08     Normalization frequency:               10
B053F09     Number of zeroes:                      2
B053F14     Number of poles:                       8
#		Complex zeroes:
#		  i  real          imag          real_error    imag_error
B053F10-13    0  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
B053F10-13    1  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
#		Complex poles:
#		  i  real          imag          real_error    imag_error
B053F15-18    0 -5.969000E-01  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    1 -5.969000E-01  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    2 -2.764600E+02  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    3 -2.764600E+02  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    4 -7.213730E+01  1.741510E+02  0.000000E+00  0.000000E+00
B053F15-18    5 -7.213730E+01 -1.741510E+02  0.000000E+00  0.000000E+00
B053F15-18    6 -1.741510E+02  7.213730E+01  0.000000E+00  0.000000E+00
B053F15-18    7 -1.741510E+02 -7.213730E+01  0.000000E+00  0.000000E+00
#		
#		+                  +---------------------------------------+                  +
#		+                  |       Channel Gain,   CSH ch EHZ      |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 2
B058F04     Gain:                                  2.433090E+03
B058F05     Frequency of gain:                     5.000000E+00 HZ
B058F06     Number of calibrations:                0
#		
#		+               +-------------------------------------------+                 +
#		+               |   Response (Coefficients),   CSH ch EHZ   |                 +
#		+               +-------------------------------------------+                 +
#		
B054F03     Transfer function type:                D
B054F04     Stage sequence number:                 3
B054F05     Response in units lookup:              V - Volts
B054F06     Response out units lookup:             COUNTS - Digital Counts
B054F07     Number of numerators:                  0
B054F10     Number of denominators:                0
#		
#		+                      +------------------------------+                       +
#		+                      |   Decimation,   CSH ch EHZ   |                       +
#		+                      +------------------------------+                       +
#		
B057F03     Stage sequence number:                 3
B057F04     Input sample rate:                     1.000000E+02
B057F05     Decimation factor:                     1
B057F06     Decimation offset:                     0
B057F07     Estimated delay (seconds):             0.000000E+00
B057F08     Correction applied (seconds):          0.000000E+00
#		
#		+                  +---------------------------------------+                  +
#		+                  |       Channel Gain,   CSH ch EHZ      |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 3
B058F04     Gain:                                  8.192000E+02
B058F05     Frequency of gain:                     1.000000E+00 HZ
B058F06     Number of calibrations:                0
#		
#		+                  +---------------------------------------+                  +
#		+                  |   Channel Sensitivity,   CSH ch EHZ   |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 0
B058F04     Sensitivity:                           1.993190E+08
B058F05     Frequency of sensitivity:              1.000000E+01 HZ
B058F06     Number of calibrations:                0
#		
#		<< IRIS SEED Reader, Release 5.4beta-USGS-20190905 >>
#		
#		======== CHANNEL RESPONSE DATA ========
B050F03     Station:     CSH
B050F16     Network:     NC
B052F03     Location:    ??
B052F04     Channel:     EHZ
B052F22     Start date:  1985,295,20:15:00
B052F23     End date:    1986,092,15:30:00
#		=======================================
#		+               +--------------------------------------------+                +
#		+               |   Response (Poles & Zeros),   CSH ch EHZ   |                +
#		+               +--------------------------------------------+                +
#		
B053F03     Transfer function type:                A [Laplace Transform (Rad/sec)]
B053F04     Stage sequence number:                 1
B053F05     Response in units lookup:              M/S - Velocity in Meters Per Second
B053F06     Response out units lookup:             V - Volts
B053F07     A0 normalization factor:               1.00285
B053F08     Normalization frequency:               10
B053F09     Number of zeroes:                      2
B053F14     Number of poles:                       2
#		Complex zeroes:
#		  i  real          imag          real_error    imag_error
B053F10-13    0  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
B053F10-13    1  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
#		Complex poles:
#		  i  real          imag          real_error    imag_error
B053F15-18    0 -5.026500E+00  3.769870E+00  0.000000E+00  0.000000E+00
B053F15-18    1 -5.026500E+00 -3.769870E+00  0.000000E+00  0.000000E+00
#		
#		+                  +---------------------------------------+                  +
#		+                  |       Channel Gain,   CSH ch EHZ      |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 1
B058F04     Gain:                                  1.000000E+02
B058F05     Frequency of gain:                     1.000000E+01 HZ
B058F06     Number of calibrations:                0
#		
#		+               +--------------------------------------------+                +
#		+               |   Response (Poles & Zeros),   CSH ch EHZ   |                +
#		+               +--------------------------------------------+                +
#		
B053F03     Transfer function type:                A [Laplace Transform (Rad/sec)]
B053F04     Stage sequence number:                 2
B053F05     Response in units lookup:              V - Volts
B053F06     Response out units lookup:             V - Volts
B053F07     A0 normalization factor:               1.01498E+14
B053F08     Normalization frequency:               10
B053F09     Number of zeroes:                      2
B053F14     Number of poles:                       8
#		Complex zeroes:
#		  i  real          imag          real_error    imag_error
B053F10-13    0  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
B053F10-13    1  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
#		Complex poles:
#		  i  real          imag          real_error    imag_error
B053F15-18    0 -5.969000E-01  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    1 -5.969000E-01  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    2 -2.764600E+02  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    3 -2.764600E+02  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    4 -7.213730E+01  1.741510E+02  0.000000E+00  0.000000E+00
B053F15-18    5 -7.213730E+01 -1.741510E+02  0.000000E+00  0.000000E+00
B053F15-18    6 -1.741510E+02  7.213730E+01  0.000000E+00  0.000000E+00
B053F15-18    7 -1.741510E+02 -7.213730E+01  0.000000E+00  0.000000E+00
#		
#		+                  +---------------------------------------+                  +
#		+                  |       Channel Gain,   CSH ch EHZ      |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 2
B058F04     Gain:                                  1.216550E+03
B058F05     Frequency of gain:                     5.000000E+00 HZ
B058F06     Number of calibrations:                0
#		
#		+               +-------------------------------------------+                 +
#		+               |   Response (Coefficients),   CSH ch EHZ   |                 +
#		+               +-------------------------------------------+                 +
#		
B054F03     Transfer function type:                D
B054F04     Stage sequence number:                 3
B054F05     Response in units lookup:              V - Volts
B054F06     Response out units lookup:             COUNTS - Digital Counts
B054F07     Number of numerators:                  0
B054F10     Number of denominators:                0
#		
#		+                      +------------------------------+                       +
#		+                      |   Decimation,   CSH ch EHZ   |                       +
#		+                      +------------------------------+                       +
#		
B057F03     Stage sequence number:                 3
B057F04     Input sample rate:                     1.000000E+02
B057F05     Decimation factor:                     1
B057F06     Decimation offset:                     0
B057F07     Estimated delay (seconds):             0.000000E+00
B057F08     Correction applied (seconds):          0.000000E+00
#		
#		+                  +---------------------------------------+                  +
#		+                  |       Channel Gain,   CSH ch EHZ      |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 3
B058F04     Gain:                                  8.192000E+02
B058F05     Frequency of gain:                     1.000000E+00 HZ
B058F06     Number of calibrations:                0
#		
#		+                  +---------------------------------------+                  +
#		+                  |   Channel Sensitivity,   CSH ch EHZ   |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 0
B058F04     Sensitivity:                           9.965980E+07
B058F05     Frequency of sensitivity:              1.000000E+01 HZ
B058F06     Number of calibrations:                0
#		
#		<< IRIS SEED Reader, Release 5.4beta-USGS-20190905 >>
#		
#		======== CHANNEL RESPONSE DATA ========
B050F03     Station:     CSH
B050F16     Network:     NC
B052F03     Location:    ??
B052F04     Channel:     EHZ
B052F22     Start date:  1986,092,15:30:00
B052F23     End date:    1986,255,21:15:00
#		=======================================
#		+               +--------------------------------------------+                +
#		+               |   Response (Poles & Zeros),   CSH ch EHZ   |                +
#		+               +--------------------------------------------+                +
#		
B053F03     Transfer function type:                A [Laplace Transform (Rad/sec)]
B053F04     Stage sequence number:                 1
B053F05     Response in units lookup:              M/S - Velocity in Meters Per Second
B053F06     Response out units lookup:             V - Volts
B053F07     A0 normalization factor:               1.00285
B053F08     Normalization frequency:               10
B053F09     Number of zeroes:                      2
B053F14     Number of poles:                       2
#		Complex zeroes:
#		  i  real          imag          real_error    imag_error
B053F10-13    0  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
B053F10-13    1  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
#		Complex poles:
#		  i  real          imag          real_error    imag_error
B053F15-18    0 -5.026500E+00  3.769870E+00  0.000000E+00  0.000000E+00
B053F15-18    1 -5.026500E+00 -3.769870E+00  0.000000E+00  0.000000E+00
#		
#		+                  +---------------------------------------+                  +
#		+                  |       Channel Gain,   CSH ch EHZ      |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 1
B058F04     Gain:                                  1.000000E+02
B058F05     Frequency of gain:                     1.000000E+01 HZ
B058F06     Number of calibrations:                0
#		
#		+               +--------------------------------------------+                +
#		+               |   Response (Poles & Zeros),   CSH ch EHZ   |                +
#		+               +--------------------------------------------+                +
#		
B053F03     Transfer function type:                A [Laplace Transform (Rad/sec)]
B053F04     Stage sequence number:                 2
B053F05     Response in units lookup:              V - Volts
B053F06     Response out units lookup:             V - Volts
B053F07     A0 normalization factor:               1.01498E+14
B053F08     Normalization frequency:               10
B053F09     Number of zeroes:                      2
B053F14     Number of poles:                       8
#		Complex zeroes:
#		  i  real          imag          real_error    imag_error
B053F10-13    0  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
B053F10-13    1  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
#		Complex poles:
#		  i  real          imag          real_error    imag_error
B053F15-18    0 -5.969000E-01  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    1 -5.969000E-01  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    2 -2.764600E+02  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    3 -2.764600E+02  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    4 -7.213730E+01  1.741510E+02  0.000000E+00  0.000000E+00
B053F15-18    5 -7.213730E+01 -1.741510E+02  0.000000E+00  0.000000E+00
B053F15-18    6 -1.741510E+02  7.213730E+01  0.000000E+00  0.000000E+00
B053F15-18    7 -1.741510E+02 -7.213730E+01  0.000000E+00  0.000000E+00
#		
#		+                  +---------------------------------------+                  +
#		+                  |       Channel Gain,   CSH ch EHZ      |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 2
B058F04     Gain:                                  1.216550E+03
B058F05     Frequency of gain:                     5.000000E+00 HZ
B058F06     Number of calibrations:                0
#		
#		+               +-------------------------------------------+                 +
#		+               |   Response (Coefficients),   CSH ch EHZ   |                 +
#		+               +-------------------------------------------+                 +
#		
B054F03     Transfer function type:                D
B054F04     Stage sequence number:                 3
B054F05     Response in units lookup:              V - Volts
B054F06     Response out units lookup:             COUNTS - Digital Counts
B054F07     Number of numerators:                  0
B054F10     Number of denominators:                0
#		
#		+                      +------------------------------+                       +
#		+                      |   Decimation,   CSH ch EHZ   |                       +
#		+                      +------------------------------+                       +
#		
B057F03     Stage sequence number:                 3
B057F04     Input sample rate:                     1.000000E+02
B057F05     Decimation factor:                     1
B057F06     Decimation offset:                     0
B057F07     Estimated delay (seconds):             0.000000E+00
B057F08     Correction applied (seconds):          0.000000E+00
#		
#		+                  +---------------------------------------+                  +
#		+                  |       Channel Gain,   CSH ch EHZ      |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 3
B058F04     Gain:                                  8.192000E+02
B058F05     Frequency of gain:                     1.000000E+00 HZ
B058F06     Number of calibrations:                0
#		
#		+                  +---------------------------------------+                  +
#		+                  |   Channel Sensitivity,   CSH ch EHZ   |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 0
B058F04     Sensitivity:                           9.965980E+07
B058F05     Frequency of sensitivity:              1.000000E+01 HZ
B058F06     Number of calibrations:                0
#		
#		<< IRIS SEED Reader, Release 5.4beta-USGS-20190905 >>
#		
#		======== CHANNEL RESPONSE DATA ========
B050F03     Station:     CSH
B050F16     Network:     NC
B052F03     Location:    ??
B052F04     Channel:     EHZ
B052F22     Start date:  1986,255,21:15:00
B052F23     End date:    1987,121,00:00:00
#		=======================================
#		+               +--------------------------------------------+                +
#		+               |   Response (Poles & Zeros),   CSH ch EHZ   |                +
#		+               +--------------------------------------------+                +
#		
B053F03     Transfer function type:                A [Laplace Transform (Rad/sec)]
B053F04     Stage sequence number:                 1
B053F05     Response in units lookup:              M/S - Velocity in Meters Per Second
B053F06     Response out units lookup:             V - Volts
B053F07     A0 normalization factor:               1.00285
B053F08     Normalization frequency:               10
B053F09     Number of zeroes:                      2
B053F14     Number of poles:                       2
#		Complex zeroes:
#		  i  real          imag          real_error    imag_error
B053F10-13    0  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
B053F10-13    1  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
#		Complex poles:
#		  i  real          imag          real_error    imag_error
B053F15-18    0 -5.026500E+00  3.769870E+00  0.000000E+00  0.000000E+00
B053F15-18    1 -5.026500E+00 -3.769870E+00  0.000000E+00  0.000000E+00
#		
#		+                  +---------------------------------------+                  +
#		+                  |       Channel Gain,   CSH ch EHZ      |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 1
B058F04     Gain:                                  1.000000E+02
B058F05     Frequency of gain:                     1.000000E+01 HZ
B058F06     Number of calibrations:                0
#		
#		+               +--------------------------------------------+                +
#		+               |   Response (Poles & Zeros),   CSH ch EHZ   |                +
#		+               +--------------------------------------------+                +
#		
B053F03     Transfer function type:                A [Laplace Transform (Rad/sec)]
B053F04     Stage sequence number:                 2
B053F05     Response in units lookup:              V - Volts
B053F06     Response out units lookup:             V - Volts
B053F07     A0 normalization factor:               1.01498E+14
B053F08     Normalization frequency:               10
B053F09     Number of zeroes:                      2
B053F14     Number of poles:                       8
#		Complex zeroes:
#		  i  real          imag          real_error    imag_error
B053F10-13    0  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
B053F10-13    1  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
#		Complex poles:
#		  i  real          imag          real_error    imag_error
B053F15-18    0 -5.969000E-01  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    1 -5.969000E-01  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    2 -2.764600E+02  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    3 -2.764600E+02  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    4 -7.213730E+01  1.741510E+02  0.000000E+00  0.000000E+00
B053F15-18    5 -7.213730E+01 -1.741510E+02  0.000000E+00  0.000000E+00
B053F15-18    6 -1.741510E+02  7.213730E+01  0.000000E+00  0.000000E+00
B053F15-18    7 -1.741510E+02 -7.213730E+01  0.000000E+00  0.000000E+00
#		
#		+                  +---------------------------------------+                  +
#		+                  |       Channel Gain,   CSH ch EHZ      |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 2
B058F04     Gain:                                  1.216550E+03
B058F05     Frequency of gain:                     5.000000E+00 HZ
B058F06     Number of calibrations:                0
#		
#		+               +-------------------------------------------+                 +
#		+               |   Response (Coefficients),   CSH ch EHZ   |                 +
#		+               +-------------------------------------------+                 +
#		
B054F03     Transfer function type:                D
B054F04     Stage sequence number:                 3
B054F05     Response in units lookup:              V - Volts
B054F06     Response out units lookup:             COUNTS - Digital Counts
B054F07     Number of numerators:                  0
B054F10     Number of denominators:                0
#		
#		+                      +------------------------------+                       +
#		+                      |   Decimation,   CSH ch EHZ   |                       +
#		+                      +------------------------------+                       +
#		
B057F03     Stage sequence number:                 3
B057F04     Input sample rate:                     1.000000E+02
B057F05     Decimation factor:                     1
B057F06     Decimation offset:                     0
B057F07     Estimated delay (seconds):             0.000000E+00
B057F08     Correction applied (seconds):          0.000000E+00
#		
#		+                  +---------------------------------------+                  +
#		+                  |       Channel Gain,   CSH ch EHZ      |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 3
B058F04     Gain:                                  8.192000E+02
B058F05     Frequency of gain:                     1.000000E+00 HZ
B058F06     Number of calibrations:                0
#		
#		+                  +---------------------------------------+                  +
#		+                  |   Channel Sensitivity,   CSH ch EHZ   |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 0
B058F04     Sensitivity:                           9.965980E+07
B058F05     Frequency of sensitivity:              1.000000E+01 HZ
B058F06     Number of calibrations:                0
#		
#		<< IRIS SEED Reader, Release 5.4beta-USGS-20190905 >>
#		
#		======== CHANNEL RESPONSE DATA ========
B050F03     Station:     CSH
B050F16     Network:     NC
B052F03     Location:    ??
B052F04     Channel:     EHZ
B052F22     Start date:  1987,121,00:00:00
B052F23     End date:    1987,161,19:55:00
#		=======================================
#		+               +--------------------------------------------+                +
#		+               |   Response (Poles & Zeros),   CSH ch EHZ   |                +
#		+               +--------------------------------------------+                +
#		
B053F03     Transfer function type:                A [Laplace Transform (Rad/sec)]
B053F04     Stage sequence number:                 1
B053F05     Response in units lookup:              M/S - Velocity in Meters Per Second
B053F06     Response out units lookup:             V - Volts
B053F07     A0 normalization factor:               1.00285
B053F08     Normalization frequency:               10
B053F09     Number of zeroes:                      2
B053F14     Number of poles:                       2
#		Complex zeroes:
#		  i  real          imag          real_error    imag_error
B053F10-13    0  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
B053F10-13    1  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
#		Complex poles:
#		  i  real          imag          real_error    imag_error
B053F15-18    0 -5.026500E+00  3.769870E+00  0.000000E+00  0.000000E+00
B053F15-18    1 -5.026500E+00 -3.769870E+00  0.000000E+00  0.000000E+00
#		
#		+                  +---------------------------------------+                  +
#		+                  |       Channel Gain,   CSH ch EHZ      |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 1
B058F04     Gain:                                  1.000000E+02
B058F05     Frequency of gain:                     1.000000E+01 HZ
B058F06     Number of calibrations:                0
#		
#		+               +--------------------------------------------+                +
#		+               |   Response (Poles & Zeros),   CSH ch EHZ   |                +
#		+               +--------------------------------------------+                +
#		
B053F03     Transfer function type:                A [Laplace Transform (Rad/sec)]
B053F04     Stage sequence number:                 2
B053F05     Response in units lookup:              V - Volts
B053F06     Response out units lookup:             V - Volts
B053F07     A0 normalization factor:               2.0085E+13
B053F08     Normalization frequency:               10
B053F09     Number of zeroes:                      2
B053F14     Number of poles:                       8
#		Complex zeroes:
#		  i  real          imag          real_error    imag_error
B053F10-13    0  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
B053F10-13    1  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
#		Complex poles:
#		  i  real          imag          real_error    imag_error
B053F15-18    0 -5.969000E-01  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    1 -5.969000E-01  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    2 -2.764600E+02  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    3 -2.764600E+02  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    4 -4.809150E+01  1.160970E+02  0.000000E+00  0.000000E+00
B053F15-18    5 -4.809150E+01 -1.160970E+02  0.000000E+00  0.000000E+00
B053F15-18    6 -1.161010E+02  4.808320E+01  0.000000E+00  0.000000E+00
B053F15-18    7 -1.161010E+02 -4.808320E+01  0.000000E+00  0.000000E+00
#		
#		+                  +---------------------------------------+                  +
#		+                  |       Channel Gain,   CSH ch EHZ      |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 2
B058F04     Gain:                                  1.216550E+03
B058F05     Frequency of gain:                     5.000000E+00 HZ
B058F06     Number of calibrations:                0
#		
#		+               +-------------------------------------------+                 +
#		+               |   Response (Coefficients),   CSH ch EHZ   |                 +
#		+               +-------------------------------------------+                 +
#		
B054F03     Transfer function type:                D
B054F04     Stage sequence number:                 3
B054F05     Response in units lookup:              V - Volts
B054F06     Response out units lookup:             COUNTS - Digital Counts
B054F07     Number of numerators:                  0
B054F10     Number of denominators:                0
#		
#		+                      +------------------------------+                       +
#		+                      |   Decimation,   CSH ch EHZ   |                       +
#		+                      +------------------------------+                       +
#		
B057F03     Stage sequence number:                 3
B057F04     Input sample rate:                     1.000000E+02
B057F05     Decimation factor:                     1
B057F06     Decimation offset:                     0
B057F07     Estimated delay (seconds):             0.000000E+00
B057F08     Correction applied (seconds):          0.000000E+00
#		
#		+                  +---------------------------------------+                  +
#		+                  |       Channel Gain,   CSH ch EHZ      |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 3
B058F04     Gain:                                  8.192000E+02
B058F05     Frequency of gain:                     1.000000E+00 HZ
B058F06     Number of calibrations:                0
#		
#		+                  +---------------------------------------+                  +
#		+                  |   Channel Sensitivity,   CSH ch EHZ   |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 0
B058F04     Sensitivity:                           9.965980E+07
B058F05     Frequency of sensitivity:              1.000000E+01 HZ
B058F06     Number of calibrations:                0
#		
#		<< IRIS SEED Reader, Release 5.4beta-USGS-20190905 >>
#		
#		======== CHANNEL RESPONSE DATA ========
B050F03     Station:     CSH
B050F16     Network:     NC
B052F03     Location:    ??
B052F04     Channel:     EHZ
B052F22     Start date:  1987,161,19:55:00
B052F23     End date:    1988,293,21:45:00
#		=======================================
#		+               +--------------------------------------------+                +
#		+               |   Response (Poles & Zeros),   CSH ch EHZ   |                +
#		+               +--------------------------------------------+                +
#		
B053F03     Transfer function type:                A [Laplace Transform (Rad/sec)]
B053F04     Stage sequence number:                 1
B053F05     Response in units lookup:              M/S - Velocity in Meters Per Second
B053F06     Response out units lookup:             V - Volts
B053F07     A0 normalization factor:               1.00285
B053F08     Normalization frequency:               10
B053F09     Number of zeroes:                      2
B053F14     Number of poles:                       2
#		Complex zeroes:
#		  i  real          imag          real_error    imag_error
B053F10-13    0  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
B053F10-13    1  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
#		Complex poles:
#		  i  real          imag          real_error    imag_error
B053F15-18    0 -5.026500E+00  3.769870E+00  0.000000E+00  0.000000E+00
B053F15-18    1 -5.026500E+00 -3.769870E+00  0.000000E+00  0.000000E+00
#		
#		+                  +---------------------------------------+                  +
#		+                  |       Channel Gain,   CSH ch EHZ      |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 1
B058F04     Gain:                                  1.000000E+02
B058F05     Frequency of gain:                     1.000000E+01 HZ
B058F06     Number of calibrations:                0
#		
#		+               +--------------------------------------------+                +
#		+               |   Response (Poles & Zeros),   CSH ch EHZ   |                +
#		+               +--------------------------------------------+                +
#		
B053F03     Transfer function type:                A [Laplace Transform (Rad/sec)]
B053F04     Stage sequence number:                 2
B053F05     Response in units lookup:              V - Volts
B053F06     Response out units lookup:             V - Volts
B053F07     A0 normalization factor:               2.0085E+13
B053F08     Normalization frequency:               10
B053F09     Number of zeroes:                      2
B053F14     Number of poles:                       8
#		Complex zeroes:
#		  i  real          imag          real_error    imag_error
B053F10-13    0  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
B053F10-13    1  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
#		Complex poles:
#		  i  real          imag          real_error    imag_error
B053F15-18    0 -5.969000E-01  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    1 -5.969000E-01  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    2 -2.764600E+02  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    3 -2.764600E+02  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    4 -4.809150E+01  1.160970E+02  0.000000E+00  0.000000E+00
B053F15-18    5 -4.809150E+01 -1.160970E+02  0.000000E+00  0.000000E+00
B053F15-18    6 -1.161010E+02  4.808320E+01  0.000000E+00  0.000000E+00
B053F15-18    7 -1.161010E+02 -4.808320E+01  0.000000E+00  0.000000E+00
#		
#		+                  +---------------------------------------+                  +
#		+                  |       Channel Gain,   CSH ch EHZ      |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 2
B058F04     Gain:                                  1.216550E+03
B058F05     Frequency of gain:                     5.000000E+00 HZ
B058F06     Number of calibrations:                0
#		
#		+               +-------------------------------------------+                 +
#		+               |   Response (Coefficients),   CSH ch EHZ   |                 +
#		+               +-------------------------------------------+                 +
#		
B054F03     Transfer function type:                D
B054F04     Stage sequence number:                 3
B054F05     Response in units lookup:              V - Volts
B054F06     Response out units lookup:             COUNTS - Digital Counts
B054F07     Number of numerators:                  0
B054F10     Number of denominators:                0
#		
#		+                      +------------------------------+                       +
#		+                      |   Decimation,   CSH ch EHZ   |                       +
#		+                      +------------------------------+                       +
#		
B057F03     Stage sequence number:                 3
B057F04     Input sample rate:                     1.000000E+02
B057F05     Decimation factor:                     1
B057F06     Decimation offset:                     0
B057F07     Estimated delay (seconds):             0.000000E+00
B057F08     Correction applied (seconds):          0.000000E+00
#		
#		+                  +---------------------------------------+                  +
#		+                  |       Channel Gain,   CSH ch EHZ      |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 3
B058F04     Gain:                                  8.192000E+02
B058F05     Frequency of gain:                     1.000000E+00 HZ
B058F06     Number of calibrations:                0
#		
#		+                  +---------------------------------------+                  +
#		+                  |   Channel Sensitivity,   CSH ch EHZ   |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 0
B058F04     Sensitivity:                           9.965980E+07
B058F05     Frequency of sensitivity:              1.000000E+01 HZ
B058F06     Number of calibrations:                0
#		
#		<< IRIS SEED Reader, Release 5.4beta-USGS-20190905 >>
#		
#		======== CHANNEL RESPONSE DATA ========
B050F03     Station:     CSH
B050F16     Network:     NC
B052F03     Location:    ??
B052F04     Channel:     EHZ
B052F22     Start date:  1988,293,21:45:00
B052F23     End date:    1989,080,20:55:00
#		=======================================
#		+               +--------------------------------------------+                +
#		+               |   Response (Poles & Zeros),   CSH ch EHZ   |                +
#		+               +--------------------------------------------+                +
#		
B053F03     Transfer function type:                A [Laplace Transform (Rad/sec)]
B053F04     Stage sequence number:                 1
B053F05     Response in units lookup:              M/S - Velocity in Meters Per Second
B053F06     Response out units lookup:             V - Volts
B053F07     A0 normalization factor:               1.00285
B053F08     Normalization frequency:               10
B053F09     Number of zeroes:                      2
B053F14     Number of poles:                       2
#		Complex zeroes:
#		  i  real          imag          real_error    imag_error
B053F10-13    0  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
B053F10-13    1  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
#		Complex poles:
#		  i  real          imag          real_error    imag_error
B053F15-18    0 -5.026500E+00  3.769870E+00  0.000000E+00  0.000000E+00
B053F15-18    1 -5.026500E+00 -3.769870E+00  0.000000E+00  0.000000E+00
#		
#		+                  +---------------------------------------+                  +
#		+                  |       Channel Gain,   CSH ch EHZ      |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 1
B058F04     Gain:                                  1.000000E+02
B058F05     Frequency of gain:                     1.000000E+01 HZ
B058F06     Number of calibrations:                0
#		
#		+               +--------------------------------------------+                +
#		+               |   Response (Poles & Zeros),   CSH ch EHZ   |                +
#		+               +--------------------------------------------+                +
#		
B053F03     Transfer function type:                A [Laplace Transform (Rad/sec)]
B053F04     Stage sequence number:                 2
B053F05     Response in units lookup:              V - Volts
B053F06     Response out units lookup:             V - Volts
B053F07     A0 normalization factor:               2.0085E+13
B053F08     Normalization frequency:               10
B053F09     Number of zeroes:                      2
B053F14     Number of poles:                       8
#		Complex zeroes:
#		  i  real          imag          real_error    imag_error
B053F10-13    0  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
B053F10-13    1  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
#		Complex poles:
#		  i  real          imag          real_error    imag_error
B053F15-18    0 -5.969000E-01  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    1 -5.969000E-01  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    2 -2.764600E+02  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    3 -2.764600E+02  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    4 -4.809150E+01  1.160970E+02  0.000000E+00  0.000000E+00
B053F15-18    5 -4.809150E+01 -1.160970E+02  0.000000E+00  0.000000E+00
B053F15-18    6 -1.161010E+02  4.808320E+01  0.000000E+00  0.000000E+00
B053F15-18    7 -1.161010E+02 -4.808320E+01  0.000000E+00  0.000000E+00
#		
#		+                  +---------------------------------------+                  +
#		+                  |       Channel Gain,   CSH ch EHZ      |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 2
B058F04     Gain:                                  1.216550E+03
B058F05     Frequency of gain:                     5.000000E+00 HZ
B058F06     Number of calibrations:                0
#		
#		+               +-------------------------------------------+                 +
#		+               |   Response (Coefficients),   CSH ch EHZ   |                 +
#		+               +-------------------------------------------+                 +
#		
B054F03     Transfer function type:                D
B054F04     Stage sequence number:                 3
B054F05     Response in units lookup:              V - Volts
B054F06     Response out units lookup:             COUNTS - Digital Counts
B054F07     Number of numerators:                  0
B054F10     Number of denominators:                0
#		
#		+                      +------------------------------+                       +
#		+                      |   Decimation,   CSH ch EHZ   |                       +
#		+                      +------------------------------+                       +
#		
B057F03     Stage sequence number:                 3
B057F04     Input sample rate:                     1.000000E+02
B057F05     Decimation factor:                     1
B057F06     Decimation offset:                     0
B057F07     Estimated delay (seconds):             0.000000E+00
B057F08     Correction applied (seconds):          0.000000E+00
#		
#		+                  +---------------------------------------+                  +
#		+                  |       Channel Gain,   CSH ch EHZ      |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 3
B058F04     Gain:                                  8.192000E+02
B058F05     Frequency of gain:                     1.000000E+00 HZ
B058F06     Number of calibrations:                0
#		
#		+                  +---------------------------------------+                  +
#		+                  |   Channel Sensitivity,   CSH ch EHZ   |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 0
B058F04     Sensitivity:                           9.965980E+07
B058F05     Frequency of sensitivity:              1.000000E+01 HZ
B058F06     Number of calibrations:                0
#		
#		<< IRIS SEED Reader, Release 5.4beta-USGS-20190905 >>
#		
#		======== CHANNEL RESPONSE DATA ========
B050F03     Station:     CSH
B050F16     Network:     NC
B052F03     Location:    ??
B052F04     Channel:     EHZ
B052F22     Start date:  1989,080,20:55:00
B052F23     End date:    1989,340,20:00:00
#		=======================================
#		+               +--------------------------------------------+                +
#		+               |   Response (Poles & Zeros),   CSH ch EHZ   |                +
#		+               +--------------------------------------------+                +
#		
B053F03     Transfer function type:                A [Laplace Transform (Rad/sec)]
B053F04     Stage sequence number:                 1
B053F05     Response in units lookup:              M/S - Velocity in Meters Per Second
B053F06     Response out units lookup:             V - Volts
B053F07     A0 normalization factor:               1.00285
B053F08     Normalization frequency:               10
B053F09     Number of zeroes:                      2
B053F14     Number of poles:                       2
#		Complex zeroes:
#		  i  real          imag          real_error    imag_error
B053F10-13    0  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
B053F10-13    1  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
#		Complex poles:
#		  i  real          imag          real_error    imag_error
B053F15-18    0 -5.026500E+00  3.769870E+00  0.000000E+00  0.000000E+00
B053F15-18    1 -5.026500E+00 -3.769870E+00  0.000000E+00  0.000000E+00
#		
#		+                  +---------------------------------------+                  +
#		+                  |       Channel Gain,   CSH ch EHZ      |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 1
B058F04     Gain:                                  1.000000E+02
B058F05     Frequency of gain:                     1.000000E+01 HZ
B058F06     Number of calibrations:                0
#		
#		+               +--------------------------------------------+                +
#		+               |   Response (Poles & Zeros),   CSH ch EHZ   |                +
#		+               +--------------------------------------------+                +
#		
B053F03     Transfer function type:                A [Laplace Transform (Rad/sec)]
B053F04     Stage sequence number:                 2
B053F05     Response in units lookup:              V - Volts
B053F06     Response out units lookup:             V - Volts
B053F07     A0 normalization factor:               2.0085E+13
B053F08     Normalization frequency:               10
B053F09     Number of zeroes:                      2
B053F14     Number of poles:                       8
#		Complex zeroes:
#		  i  real          imag          real_error    imag_error
B053F10-13    0  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
B053F10-13    1  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
#		Complex poles:
#		  i  real          imag          real_error    imag_error
B053F15-18    0 -5.969000E-01  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    1 -5.969000E-01  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    2 -2.764600E+02  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    3 -2.764600E+02  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    4 -4.809150E+01  1.160970E+02  0.000000E+00  0.000000E+00
B053F15-18    5 -4.809150E+01 -1.160970E+02  0.000000E+00  0.000000E+00
B053F15-18    6 -1.161010E+02  4.808320E+01  0.000000E+00  0.000000E+00
B053F15-18    7 -1.161010E+02 -4.808320E+01  0.000000E+00  0.000000E+00
#		
#		+                  +---------------------------------------+                  +
#		+                  |       Channel Gain,   CSH ch EHZ      |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 2
B058F04     Gain:                                  1.216550E+03
B058F05     Frequency of gain:                     5.000000E+00 HZ
B058F06     Number of calibrations:                0
#		
#		+               +-------------------------------------------+                 +
#		+               |   Response (Coefficients),   CSH ch EHZ   |                 +
#		+               +-------------------------------------------+                 +
#		
B054F03     Transfer function type:                D
B054F04     Stage sequence number:                 3
B054F05     Response in units lookup:              V - Volts
B054F06     Response out units lookup:             COUNTS - Digital Counts
B054F07     Number of numerators:                  0
B054F10     Number of denominators:                0
#		
#		+                      +------------------------------+                       +
#		+                      |   Decimation,   CSH ch EHZ   |                       +
#		+                      +------------------------------+                       +
#		
B057F03     Stage sequence number:                 3
B057F04     Input sample rate:                     1.000000E+02
B057F05     Decimation factor:                     1
B057F06     Decimation offset:                     0
B057F07     Estimated delay (seconds):             0.000000E+00
B057F08     Correction applied (seconds):          0.000000E+00
#		
#		+                  +---------------------------------------+                  +
#		+                  |       Channel Gain,   CSH ch EHZ      |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 3
B058F04     Gain:                                  8.192000E+02
B058F05     Frequency of gain:                     1.000000E+00 HZ
B058F06     Number of calibrations:                0
#		
#		+                  +---------------------------------------+                  +
#		+                  |   Channel Sensitivity,   CSH ch EHZ   |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 0
B058F04     Sensitivity:                           9.965980E+07
B058F05     Frequency of sensitivity:              1.000000E+01 HZ
B058F06     Number of calibrations:                0
#		
#		<< IRIS SEED Reader, Release 5.4beta-USGS-20190905 >>
#		
#		======== CHANNEL RESPONSE DATA ========
B050F03     Station:     CSH
B050F16     Network:     NC
B052F03     Location:    ??
B052F04     Channel:     EHZ
B052F22     Start date:  1989,340,20:00:00
B052F23     End date:    1991,056,22:03:00
#		=======================================
#		+               +--------------------------------------------+                +
#		+               |   Response (Poles & Zeros),   CSH ch EHZ   |                +
#		+               +--------------------------------------------+                +
#		
B053F03     Transfer function type:                A [Laplace Transform (Rad/sec)]
B053F04     Stage sequence number:                 1
B053F05     Response in units lookup:              M/S - Velocity in Meters Per Second
B053F06     Response out units lookup:             V - Volts
B053F07     A0 normalization factor:               1.00285
B053F08     Normalization frequency:               10
B053F09     Number of zeroes:                      2
B053F14     Number of poles:                       2
#		Complex zeroes:
#		  i  real          imag          real_error    imag_error
B053F10-13    0  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
B053F10-13    1  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
#		Complex poles:
#		  i  real          imag          real_error    imag_error
B053F15-18    0 -5.026500E+00  3.769870E+00  0.000000E+00  0.000000E+00
B053F15-18    1 -5.026500E+00 -3.769870E+00  0.000000E+00  0.000000E+00
#		
#		+                  +---------------------------------------+                  +
#		+                  |       Channel Gain,   CSH ch EHZ      |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 1
B058F04     Gain:                                  1.000000E+02
B058F05     Frequency of gain:                     1.000000E+01 HZ
B058F06     Number of calibrations:                0
#		
#		+               +--------------------------------------------+                +
#		+               |   Response (Poles & Zeros),   CSH ch EHZ   |                +
#		+               +--------------------------------------------+                +
#		
B053F03     Transfer function type:                A [Laplace Transform (Rad/sec)]
B053F04     Stage sequence number:                 2
B053F05     Response in units lookup:              V - Volts
B053F06     Response out units lookup:             V - Volts
B053F07     A0 normalization factor:               2.0085E+13
B053F08     Normalization frequency:               10
B053F09     Number of zeroes:                      2
B053F14     Number of poles:                       8
#		Complex zeroes:
#		  i  real          imag          real_error    imag_error
B053F10-13    0  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
B053F10-13    1  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
#		Complex poles:
#		  i  real          imag          real_error    imag_error
B053F15-18    0 -5.969000E-01  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    1 -5.969000E-01  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    2 -2.764600E+02  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    3 -2.764600E+02  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    4 -4.809150E+01  1.160970E+02  0.000000E+00  0.000000E+00
B053F15-18    5 -4.809150E+01 -1.160970E+02  0.000000E+00  0.000000E+00
B053F15-18    6 -1.161010E+02  4.808320E+01  0.000000E+00  0.000000E+00
B053F15-18    7 -1.161010E+02 -4.808320E+01  0.000000E+00  0.000000E+00
#		
#		+                  +---------------------------------------+                  +
#		+                  |       Channel Gain,   CSH ch EHZ      |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 2
B058F04     Gain:                                  1.216550E+03
B058F05     Frequency of gain:                     5.000000E+00 HZ
B058F06     Number of calibrations:                0
#		
#		+               +-------------------------------------------+                 +
#		+               |   Response (Coefficients),   CSH ch EHZ   |                 +
#		+               +-------------------------------------------+                 +
#		
B054F03     Transfer function type:                D
B054F04     Stage sequence number:                 3
B054F05     Response in units lookup:              V - Volts
B054F06     Response out units lookup:             COUNTS - Digital Counts
B054F07     Number of numerators:                  0
B054F10     Number of denominators:                0
#		
#		+                      +------------------------------+                       +
#		+                      |   Decimation,   CSH ch EHZ   |                       +
#		+                      +------------------------------+                       +
#		
B057F03     Stage sequence number:                 3
B057F04     Input sample rate:                     1.000000E+02
B057F05     Decimation factor:                     1
B057F06     Decimation offset:                     0
B057F07     Estimated delay (seconds):             0.000000E+00
B057F08     Correction applied (seconds):          0.000000E+00
#		
#		+                  +---------------------------------------+                  +
#		+                  |       Channel Gain,   CSH ch EHZ      |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 3
B058F04     Gain:                                  8.192000E+02
B058F05     Frequency of gain:                     1.000000E+00 HZ
B058F06     Number of calibrations:                0
#		
#		+                  +---------------------------------------+                  +
#		+                  |   Channel Sensitivity,   CSH ch EHZ   |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 0
B058F04     Sensitivity:                           9.965980E+07
B058F05     Frequency of sensitivity:              1.000000E+01 HZ
B058F06     Number of calibrations:                0
#		
#		<< IRIS SEED Reader, Release 5.4beta-USGS-20190905 >>
#		
#		======== CHANNEL RESPONSE DATA ========
B050F03     Station:     CSH
B050F16     Network:     NC
B052F03     Location:    ??
B052F04     Channel:     EHZ
B052F22     Start date:  1991,056,22:03:00
B052F23     End date:    1992,031,17:15:00
#		=======================================
#		+               +--------------------------------------------+                +
#		+               |   Response (Poles & Zeros),   CSH ch EHZ   |                +
#		+               +--------------------------------------------+                +
#		
B053F03     Transfer function type:                A [Laplace Transform (Rad/sec)]
B053F04     Stage sequence number:                 1
B053F05     Response in units lookup:              M/S - Velocity in Meters Per Second
B053F06     Response out units lookup:             V - Volts
B053F07     A0 normalization factor:               1.00285
B053F08     Normalization frequency:               10
B053F09     Number of zeroes:                      2
B053F14     Number of poles:                       2
#		Complex zeroes:
#		  i  real          imag          real_error    imag_error
B053F10-13    0  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
B053F10-13    1  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
#		Complex poles:
#		  i  real          imag          real_error    imag_error
B053F15-18    0 -5.026500E+00  3.769870E+00  0.000000E+00  0.000000E+00
B053F15-18    1 -5.026500E+00 -3.769870E+00  0.000000E+00  0.000000E+00
#		
#		+                  +---------------------------------------+                  +
#		+                  |       Channel Gain,   CSH ch EHZ      |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 1
B058F04     Gain:                                  1.000000E+02
B058F05     Frequency of gain:                     1.000000E+01 HZ
B058F06     Number of calibrations:                0
#		
#		+               +--------------------------------------------+                +
#		+               |   Response (Poles & Zeros),   CSH ch EHZ   |                +
#		+               +--------------------------------------------+                +
#		
B053F03     Transfer function type:                A [Laplace Transform (Rad/sec)]
B053F04     Stage sequence number:                 2
B053F05     Response in units lookup:              V - Volts
B053F06     Response out units lookup:             V - Volts
B053F07     A0 normalization factor:               2.0085E+13
B053F08     Normalization frequency:               10
B053F09     Number of zeroes:                      2
B053F14     Number of poles:                       8
#		Complex zeroes:
#		  i  real          imag          real_error    imag_error
B053F10-13    0  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
B053F10-13    1  0.000000E+00  0.000000E+00  0.000000E+00  0.000000E+00
#		Complex poles:
#		  i  real          imag          real_error    imag_error
B053F15-18    0 -5.969000E-01  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    1 -5.969000E-01  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    2 -2.764600E+02  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    3 -2.764600E+02  0.000000E+00  0.000000E+00  0.000000E+00
B053F15-18    4 -4.809150E+01  1.160970E+02  0.000000E+00  0.000000E+00
B053F15-18    5 -4.809150E+01 -1.160970E+02  0.000000E+00  0.000000E+00
B053F15-18    6 -1.161010E+02  4.808320E+01  0.000000E+00  0.000000E+00
B053F15-18    7 -1.161010E+02 -4.808320E+01  0.000000E+00  0.000000E+00
#		
#		+                  +---------------------------------------+                  +
#		+                  |       Channel Gain,   CSH ch EHZ      |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 2
B058F04     Gain:                                  1.216550E+03
B058F05     Frequency of gain:                     5.000000E+00 HZ
B058F06     Number of calibrations:                0
#		
#		+               +-------------------------------------------+                 +
#		+               |   Response (Coefficients),   CSH ch EHZ   |                 +
#		+               +-------------------------------------------+                 +
#		
B054F03     Transfer function type:                D
B054F04     Stage sequence number:                 3
B054F05     Response in units lookup:              V - Volts
B054F06     Response out units lookup:             COUNTS - Digital Counts
B054F07     Number of numerators:                  0
B054F10     Number of denominators:                0
#		
#		+                      +------------------------------+                       +
#		+                      |   Decimation,   CSH ch EHZ   |                       +
#		+                      +------------------------------+                       +
#		
B057F03     Stage sequence number:                 3
B057F04     Input sample rate:                     1.000000E+02
B057F05     Decimation factor:                     1
B057F06     Decimation offset:                     0
B057F07     Estimated delay (seconds):             0.000000E+00
B057F08     Correction applied (seconds):          0.000000E+00
#		
#		+                  +---------------------------------------+                  +
#		+                  |       Channel Gain,   CSH ch EHZ      |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 3
B058F04     Gain:                                  8.192000E+02
B058F05     Frequency of gain:                     1.000000E+00 HZ
B058F06     Number of calibrations:                0
#		
#		+                  +---------------------------------------+                  +
#		+                  |   Channel Sensitivity,   CSH ch EHZ   |                  +
#		+                  +---------------------------------------+                  +
#		
B058F03     Stage sequence number:                 0
B058F04     Sensitivity:                           9.965980E+07
B058F05     Frequency of sensitivity:              1.000000E+01 HZ
B058F06     Number of calibrations:                0
#		
